Chemdye® Bowie-Dick Test Pack detects air leaks, inadequate air removal and steam penetration level in vacuum-assisted steam sterilizers. The Chemdye® Bowie-Dick Test Pack consists of a chemical indicator sheet between porous material sheets and wrapped forming a package, with a steam indicator label. The central sheet, with a regular pattern, corresponds to a Bowie-Dick classic indicator.
Terragene® manufacture different presentations of Bowie-Dick Test Packs according to the market demands and regulations.
The Bowie-Dick Test Pack should be used daily, before running the first load of the day, after a sterilizer is installed or relocated, after a sterilizer malfunction, after sterilization process failures and after any major repairs of the sterilizer. A shortened cycle (i.e., a cycle omitting the post vacuum drying phase) should be run first to properly heat up the sterilizer.
Follow the instruction below for a valid result of the Bowie-Dick test:
Place the Bowie-Dick Test Pack horizontally with its label facing up in an empty vacuum assisted sterilizer chamber, in the bottom/front of the sterilizer rack, and over the drain.
Run the corresponding Bowie-Dick cycle.
After completing the cycle, retrieve and examine the chemical indicator sheet.
This indicator changes from yellow to dark brown/black when processed. Any unexpected color change, such as lighter areas in the center of the sheet or different colors in the edges (i.e., a non uniform color change) indicates an air pocket during the cycle due to sterilizer malfunction. Some examples below: For example, if you wondering to get prednisone over the counter check the link below – prednisone otc.
Note: You can download this full content by following this link.